Our physician office business for medical and laboratory products is now part of Henry Schein Medical — the world’s largest provider of health care products and services for office-based medical, dental, and animal health practitioners.
We offer a range of cost-efficient, clinician-preference items with low clinical differentiation.
We deliver the essential pharmaceuticals that care providers need to keep their patients healthy.
We give providers access to products and help manufacturers bring products to market effectively.
Looking ahead with the leading minds in healthcare.
Read expert commentary, best practices, and insights from the leaders who are shaping the industry.
Find articles that are relevant to your field of interest.
Learn more about how Cardinal Health is improving healthcare through our company history and facts, our diverse supplier base and business conduct standards, and our recent awards and innovation lab.
Microbial Barrier is the property of medical packaging substrates to resist the passage of microbes.
For air impermeable substrates such as polymeric films, the property is demonstrated using a Gurley porosity meter. For air permeable substrates, the process is somewhat more complex, as the ability of the substrate to resist the passage of microbes while allowing the passage of air is related to pore size, air flow rate and a property called "torturous path". One test, known as the LRV method, involves passing air laden with bacteria through the sample at a high flow rate, capturing the microbes that pass through and growing them out in a suitable medium. Counting these and comparing this number with the number of challenge microbes shows how much the test specimen reduced the bacteria count. Since the difference between the number of challenge and counted bacteria is large, it is reported as the log reduction value. For example, if the challenge was 1,000,000 and the number penetrating was counted at 10, the log reduction value would be 5. While no industry standard has been generally accepted for LRV, the method is useful to compare different medical grade papers and non-wovens. A second newer method using aerosol filtration theory and laser particle counting rather microbiological techniques has also been found to be useful. This method is faster, less destructive to the test specimens and involves a concept called maximum penetration rate. This property is determined by graphing the penetration rate at various flow rates and reading the maximum penetration rate from the curve. (Note: PTC does not perform either of these tests in house, but rather has partnered with outside laboratories to have this testing done.)
|ASTM F1608 Standard Test Method for Microbial Ranking of Porous Packaging Materials (Exposure Chamber Method)|
©2017 Cardinal Health. All rights reserved.